Manual autoclave with vertical loading; gravity displacement type with possibility for Vacuum.
Capacity: Up to 12 kg/h
Chamber size: >150 liters
Material: Stainless steel
Temperature: 134 °C
Minimum autoclave chamber size shall be 150 liters and shall have a usable volume (loading basket) of 100 liters.
Shall work according to gravity air displacement principle.
The vacuum system shall be of a thermal shock type.
The pressure vessels should be pressure tight, the rising of the pressure (at a start pressure of below 80 mbar)
shall be lower than 1 mbar/min.
With in-build electrical heating elements for steam generation (heating elements > 4000 Kw).
Chamber and door shall be made from high quality stainless steel (e.g. AISI 3161).
Chamber in vertical position, it shall allow an easy loading/unloading of the loading basket.
2 Baskets of >100 l for the loading and unloading of waste shall be included. The baskets shall be designed for the
usage of non-heat resistant waste bags, a system for the collection and controlled release of free liquids and
condensate from the waste shall be incorporated.
The temperature measured at the reference measurement point of the treatment chamber shall not be lower than 105°C
(TDes 0+4). The treatment time shall be ≥30 minutes at 105°C, ≥20 minutes with a temperature used of ≥121°C (TDes
0+4) and ≥10 minutes with a treatment temperature of ≥ 134.
The steering of the treatment process should be semi-automatically as well as manual.
The Treatment Equipment shall be equipped with at least the following instruments:
o treatment chamber temperature indicating instrument
otreatment chamber pressure indicating instrument
For the treatment of the outgoing air, a HEPA filter in combination with a condensation system should be used.
The filter shall be built in a way that a disinfection of the filter shall be possible in a built-in condition.
The filters must be heat resistant, steam resistant, hydrophobic and pressure resistant for the pressure difference.
Cooling down phase after treatment should be existent; the temperature of the waste should be not higher than 80°C
after finishing the cooling down process.
The pressure vessel should be equipped with a heat insulation, the maximum outer surface temperature should be < 50°